Precautions of the device
1.Operating temperature is better kept at a constant temperature of 22 degree Celsius or above.
2.Please shut down the device after using.
3.Please keep the device clean especially the card slot otherwise it might effect the results.
Range of value: 0.5-50
Data judgment standard: Proceed the first insemination at 10. Proceed the second insemination after 24 hours.Operation Steps
1. Turn on the power and turn on the orange switch on the top of the machine.
2. After the instrument performs self-test, enter the main page and click the second item, "Standard Test".
3. Use a pipette to draw 80ul of serum and drop it into the round hole of the reagent card until the serum flows into the detection zone.
4. Insert the reagent card into the machine card slot with the round hole facing forward.
5. Click "Standard Test" on the screen, or click the blue button directly to automatically count down for 15 minutes.
6. Automatically display the results and it can be printed.
Fluorescence Immunochromato Graphy
When using a 80 µl pipette:0.1 - 50 ng/ml
Run Time:15 minutes